I was using PBP3 and a 3rd party programmer just fine. Then I got labX1. I was playing around with that. Everything was working fine including the U2 programmer. I only used it once.
also can use an output driving a Mosfet ( w/ a little cap between gate and ground ) that clamps the reset pin to gnd ...
not very smart, but damn efficient !
The software reset is not a complete hardware reset. Keep that in mind, since some registers are not really reset as in a power cycle.
Yes, that's exactly what it is. The reason for the results have been covered in that thread but not a way to end up with a correct pressure reading - which is still kind of hit and miss since we need...
this is what I mean
'* Notes : For a 18F1320 and 54AL07H2210 *
was it not?
that makes it a "54A L07H2210" not what was posted , its best to be sure
steve can you read the characteristics in the data sheet ,that what comes out garbled for me
PDF works for me.
The sensor is a 0.07 psi low voltage range with output from 0.5-4.5V
Is this not a repeat of the left and right output justification as on the previous thread in Richard's link.
7040 left justified is 110 right justified using Henrik's 4.88mV then
ditto for the pdf
do we have a 54A L07H2210 or a 54ALL07H2210 or a 54AL007H2210 or a 54A 007H2210 ?
they are all different animals and the 54ALxxx are 3.3v devices